SMOOTHSKIN PURE is fitted with skin contact sensors that check your skin contact before each flash, making sure the device is safe to operate. The sensors must be in full contact with the treatment area for the device to flash.
If the device is not in full contact with the treatment area, the device will not fire. Try adjusting the positioning of the device to ensure full contact is achieved, particularly on more awkward areas like arms and shins.
Power Mode is the highest power setting of the device. Power Mode will deliver the best results, as it is using the maximum power output possible for your unique skin tone. This mode works particularly well on stubborn areas where the hair tends to be denser, such as underarms and bikini line.
Gentle Mode can be used if you find Power Mode uncomfortable or a little too painful. Operating the device in Gentle Mode will reduce the power output, and you will notice less Power Bar indicators illuminated when applied to your skin.
Speed Mode is the fastest mode, and therefore the lowest power mode. Speed Mode is best for treating larger areas where the hair is less dense (such as arms or legs), and where you want to move along the treatment area quickly. It can also be used as a maintenance mode, to be used to keep hair regrowth at bay after your initial treatment regime.
SMOOTHSKIN PURE will display any error messages through combinations of lights on the power bars on the sides of the handset. This indicates that there is a fault with the device. Stop using the device and contact us.
The smell of burning when in use can indicate that the treatment area hasn’t been properly prepared for treatment. Skin must be completely hair-free (for best results by shaving, if hair is not completely removed it may damage the front of the device), cleaned and dried. If any visible hair remains above the surface of the skin, it can burn on treatment with your SmoothSkin device. If you are concerned STOP treatment and contact us.
Carefully wipe with a soft, dry cloth and ensure that the glass filter is undamaged after every use. Please see your user guide for further guidance.
No, there should not be any glass against the skin. The glass filter is recessed back into the device.
This is perfectly normal, the lamp is quality engineered and has passed the highest standards of quality checks. This does not affect the power output.
The area you are trying to treat is too dark for treatment. SMOOTHSKIN PURE includes an intelligent skintone sensor that automatically detects your skin tone and selects the optimal treatment setting. Due to the nature of Intense Pulsed Light technology, SMOOTHSKIN PURE is not for everyone and cannot treat very dark skin.